Lang Chinese (Traditional)
LangId zh-TW
4TensTile TENS
4NmesTile NMES
4FesTile FES
AnyCustom_Prog Custom
AnyCustom_Prog_fullname Custom Programme
AnyCustom_Prog_more This programme allows you to customise your own treatment. You can create treatments ranging from 1 to 5 phases.\nPhases are useful to specify portions of a programme with a specific time duration and type. For example, you can create a 10-minute warm-up with continuous stimulation (phase 1) followed by an EMG game for voluntary training (phase 2).
AnyCustom_Prog_start 1. \\>Make sure the electrodes are suitably placed and well connected before you begin.\\<\n2. \\>Press the START button to begin the first phase and follow the on-screen prompts. You can check the progress of your programme using information on the top header of your screen.\\<\n3. \\>Use the on-screen buttons on the left to skip phases or to return to completed phases.\\<
AnyCustom_Prog_enc Please consider a suitable time schedule for your treatment, so that you can work consistently to achieve your goal.
AnyCustom_Prog_mod Multi-phase customisable programme.
PelvicFloor6 Pelvic Floor (6)
PelvicFloor22 Pelvic Floor (22)
PelvicFloor 骨盆肌
Rehab25 Rehab (25)
Rehab Rehab
Sports Sports
EMGCustom EMG治療模式
TENScustom_P1 TENS Custom
TENScustom_P1_fullname List of 10 custom TENS programmes
TENScustom_P1_more This is a set of up to 10 programmes which you can customise for Pain relief (TENS).\nEach programme can have up to 5 phases of the following phase types:\n- Continuous Stimulation (STIM CON)\n- Modulated Stimulation (STIM MOD)
TENScustom_P1_start 1. \\>Make sure the electrodes are suitably placed and well connected before you begin.\\<\n2. \\>Press the START button to begin the first phase and follow the on-screen prompts. You can check the progress of your programme using information on the top header of your screen.\\<\n3. \\>Use the on-screen buttons on the left to skip phases or to return to completed phases.\\<
TENScustom_P1_enc Please consider a suitable time schedule for your treatment, so that you can work consistently to achieve your goal.
TENScustom_P1_mod Multi-phase customisable programme.
NMEScustom_P1 NMES Custom
NMEScustom_P1_fullname List of 10 custom NMES programmes
NMEScustom_P1_more This is a set of up to 10 Programmes which you can customise for Muscle Stimulation (NMES).\nEach programme can have up to 5 phases of the following phase types:\n- Continuous Stimulation (STIM CON)\n- Modulated Stimulation (STIM MOD)\n- Synchronous Stimulation (STIM SYN)\n- Alternating Stimulation (STIM ALT).
NMEScustom_P1_start 1. \\>Make sure the electrodes are suitably placed and well connected before you begin.\\<\n2. \\>Press the START button to begin the first phase and follow the on-screen prompts. You can check the progress of your\\<
NMEScustom_P1_enc Please consider a suitable time schedule for your treatment, so that you can work consistently to achieve your goal.
NMEScustom_P1_mod Multi-phase customisable programme.
FEScustom_P1 FES Custom
FEScustom_P1_fullname List of 10 custom FES programmes
FEScustom_P1_more This is a set of up to 10 Programmes which you can customise for Functional electro-stimulation (FES).\nEach programme can have up to 5 phases of the following phase types:\n- Continuous Stimulation (STIM CON)\n- Modulated Stimulation (STIM MOD)\n- Synchronous Stimulation (STIM SYN)\n- Alternating Stimulation (STIM ALT)\n- Electromyography (EMG)\n- EMG triggered STIM (ETS).
FEScustom_P1_start 1. \\>Make sure the electrodes are suitably placed and well connected before you begin.\\<\n2. \\>Press START button to begin the first phase and follow the on-screen prompts. You can check the progress of your programme using information on the top header of your screen.\\<\n3. \\>Use the on-screen buttons on the left to skip phases or to return to completed phases.\\<
FEScustom_P1_enc Please consider a suitable time schedule for your treatment, so that you can work consistently to achieve your goal.
FEScustom_P1_mod Multi-phase customisable programme.
EMGCustom_P1 EMG值
EMGCustom_P1_fullname EMG值
EMGCustom_P1_more EMG代表肌電圖,亦被稱為肌肉回饋，可以透過螢幕反映看到肌肉的性能。EMG治療有兩種目的：一，對於肌肉自主收縮練習來說，它是很有用的工具。二，如果你需要評估肌肉修復進度，EMG也是非常有用的評估工具。
EMGCustom_P1_start 1. \\>EMG治療的設定：按“SET”鍵5秒，進入“段”設定。按“SELECT PARAM”鍵來進行滾動設置，從而調節參數數值。按“ECS”鍵退出設置。 \\<\n2. \\>確定你要測量的肌肉。根據電極片位置粘貼圖表把電極片粘貼在身體上。對於較大的肌肉組織要使用大尺寸的電極片。把兩個通道的電極片都粘貼在將要被測量的肌肉上，外加參考電極片。每個通道有兩個電極片，其中一個電極的電極片應該粘貼在肌肉的運動點上。運動點一般在肌肉的中心，也就是運動神經進入肌肉的地方。另一通道則粘貼在靠近肌肉的上或頂部位置。電極片之間的距離應不少於1cm。 \\<\n3. \\>為了獲得精確的EMG測量值，請使用(REF)輔助線。 \\<\n4. \\>粘貼好電極片後，就進入EMG圖介面，試著放鬆肌肉王參數值低於4μV。若讀數不穩定，而圖表讀數沒有下降，請確保：a) 臨界值已設定為10μV或更低 b) 已使用輔助線 c）已遠離電極磁場：如行動電話、個人電腦、電源設備。 \\<\n5. \\>當你能放鬆至4μV以下，便可以開始運行EMG治療。 \\<\n6. \\>按“START”鍵，然後跟著提示操作。當你看到向上箭嘴閃動時，請收縮肌肉直至看到放鬆提示-箭嘴向下閃動。收縮肌肉時，盡力讓你的EMG圖表值維持在臨界值線上。放鬆時，盡可能快速地放鬆。能放鬆至小於4μV是不錯的表現。這種“收縮-放鬆”的迴圈將重複出現很多次，參照設置。 \\<\n7. \\>EMG治療結束後，統計資料會顯示在螢幕上。\\<
EMGCustom_P1_enc 
EMGCustom_P1_mod This programme allows you to customise your own EMG treatment.
STIMCustom STIM治療模式
STIMCustom_P1 STIM
STIMCustom_P1_fullname STIM
STIMCustom_P1_more STIM 相等於刺激。當肌肉受一系列電極脈衝刺激而收縮時，稱作NMS(神經肌肉刺激)。STIM可用於廣泛的理料用途：肌肉修復、肌肉鍛煉、為提升自控能力的骨盆鍛鍊。
STIMCustom_P1_start 1. \\>設定STIM治療：按"SET"鍵5秒，以進入段設置。選擇段類型：STIM CON類型, SYN類型, ALT 類型或MOD類型，然後按“SELECT PARAM”鍵來進行設置，以調節參數值。設定了段後，按“SET”鍵來設定下一個段。若要刪除段，只需把段時間設置為0秒。按“ECS”鍵退出設置介面。 \\<\n2. \\>確定你要刺激的肌肉。根據電極片位置粘貼圖表把電極片粘貼在身體上。對於較大的肌肉組織要使用大尺寸的電極片。你把兩個通道的電極片粘貼在將要被測量的肌肉上，外加參考電極片。每個通道有兩個電極片，其中一個電極的電極片應該粘貼在肌肉的運動點上。運動點一般在肌肉的中心，也就是運動神經進入肌肉的地方。另一通道則粘貼在靠近肌肉的上或頂部位置。電極片之間的距離應不少於1cm。 \\<\n3. \\>調節電流(mA)，以展開STIM治療。\\<
STIMCustom_P1_enc 
STIMCustom_P2 
ETSCustom ETS治療模式
ETSCustom_P1 ETS
ETSCustom_P1_fullname ETS
ETSCustom_P1_more ETS是指由電刺激而觸發的EMG。使用ETS治療模式前，可先使用EMG治療。你會發現ETS和EMG很相似。只有一點本質性的區別：在ETS模式下，當你收縮肌肉至臨界值時，會觸發電刺激，這些電刺激會說明肌肉維持收縮狀態。ETS對肌肉修復有顯著的功效，特別是針對尿失禁和中風的治療。
ETSCustom_P1_start 1. \\>EMG治療的設定：按“SET”鍵5秒，進入“段”設定。按“SELECT PARAM”鍵來進行設置，從而調節參數數值。按“ECS”鍵退出設置。 \\<\n2. \\>確定你要測量的肌肉。根據電極片位置粘貼圖表把電極片粘貼在身體上。對於較大的肌肉組織要使用大尺寸的電極片。把兩個通道的電極片都粘貼在將要被測量的肌肉上，外加參考電極片。每個通道有兩個電極片，其中一個電極的電極片應該粘貼在肌肉的運動點上。運動點一般在肌肉的中心，也就是運動神經進入肌肉的地方。另一通道則粘貼在靠近肌肉的上或頂部位置。電極片之間的距離應不少於1cm。 \\<\n3. \\>為了獲得精確的EMG測量值，請使用(REF)輔助線。 \\<\n4. \\>粘貼好電極片後，就進入EMG圖介面，試著放鬆肌肉王參數值低於4uV。若讀數不穩定，而圖表讀數沒有下降，請確保：a) 臨界值已設定為10uV或更低 b) 已使用輔助線 c）已遠離電極磁場：如行動電話、個人電腦、電源設備。 \\<\n5. \\>當你能放鬆至4uV以下，便可以開始運行EMG治療。 \\<\n6. \\>按“START”鍵，然後跟著提示操作。當你看到向上箭嘴閃動時，請收縮肌肉直至看到放鬆提示-箭嘴向下閃動。收縮肌肉時，盡力讓你的EMG圖表值維持在臨界值線上。放鬆時，盡可能快速地放鬆。能放鬆至小於4uV是不錯的表現。這種“收縮-放鬆”的迴圈將重複出現很多次，參照設置。 \\<\n7. \\>EMG治療結束後，統計資料會顯示在螢幕上。\\<
ETSCustom_P1_enc 
ETSCustom_P1_mod This programme allows you to customise your own ETS treatment.
ETSCustom_P2 
ETSCustom_P2_fullname 
Template_P1 Template 10/6
Template_P1_fullname Template: 10 sec of relaxation / 6 sec of contraction
Template_P1_more This template is ideal to get started with this mode of treatment. Once this template is mastered, you can use other more demanding templates.
Template_P1_start Press the on-screen START button and follow the template by contracting and relaxing. Try to keep your EMG levels (red) within the yellow template line, or as close to it as possible.
Template_P1_enc Use this template on a daily basis for several months and then twice a week as a follow-up to help keep you in shape.
Template_P1_mod EMG Biofeedback Template
Template_P2 Template 10/3-3-3
Template_P2_fullname Template: 10 sec. of relaxation, 3 rapid contractions of 3 seconds each
Template_P2_more This training can be used for training of fast twitch fibres.
Template_P2_start Press the on-screen START button and follow the template by contracting and relaxing. Try to keep your EMG levels (red) within the yellow template line, or as close to it as possible
Template_P2_enc Use this template on a daily basis for several months and then twice a week as a follow-up to help keep you in shape.
Template_P2_mod EMG Biofeedback Template
Template_P3 Template 10/8
Template_P3_fullname Template: 10 sec. of relaxation, 8 sec of contraction
Template_P3_more This training can be used for general training of slow twitch fibers once the patient’s condition has improved. It may be too difficult at an early stage of rehabilitation – if so, please start from Template 1: 10/6.
Template_P3_start Press the on-screen START button and follow the template by contracting and relaxing. Try to keep your EMG levels (red) within the yellow template line, or as close to it as possible.
Template_P3_enc Use this template on a daily basis for several months and then twice a week as a follow-up to help keep you in shape.
Template_P3_mod EMG Biofeedback Template
Template_P4 Template 10/10
Template_P4_fullname Template: 10 sec of relaxation, 10 sec of contraction
Template_P4_more This training can be used for general training of slow twitch fibers once the patient’s condition has improved. It may be too difficult at an early stage of rehabilitation – if so, please start from Template 1: 10/6.
Template_P4_start Press the on-screen START button and follow the template by contracting and relaxing. Try to keep your EMG levels (red) within the yellow template line, or as close to it as possible.
Template_P4_enc Use this template on a daily basis for several months and then twice a week as a follow-up to help keep you in shape.
Template_P4_mod EMG Biofeedback Template
Template_P5 Custom Shape
Template_P5_fullname Custom Shape
Template_P5_more This programme allows you to draw your own custom Template line. For example, you can draw a relaxation curve which gradually decreases over time.
Template_P5_start Press the on-screen START button and follow the template by contracting and relaxing. Try to keep your EMG levels (red) within the yellow template line, or as close to it as possible.
Template_P5_enc Use this template on a daily basis for several months and then twice a week as a follow-up to help keep you in shape.
Template_P5_mod EMG Biofeedback Template
4PelvicTile Pelvic floor
PelvicFloor_P1 骨盆底疼痛
PelvicFloor_P1_fullname 骨盆底疼痛
PelvicFloor_P1_more 骨盆底疼痛的原因有很多，並常與其它疾病相伴而生，例如：膀胱炎或腸道炎、便秘、泌尿系統失調。復發的非傳染性炎症、陰道痙攣、尾椎骨或骶髂關節炎症也會導致骨盆底疼痛。這些病症亦會引發骨盆肌肉張力增加和血液迴圈等問題。
PelvicFloor_P1_start 使用說明： 1.你可以使用陰道或直腸探針或把電極片粘貼在脊椎下部的腰骶骨區。你可使用單通道或雙通道進行電刺激。 2.如果使用陰道探針，輕輕把乾淨且濕潤的電極探針放入陰道。請確保探針的金屬部分完全放入陰道。探針的金屬表面朝向大腿，探針的表面要與陰道壁接觸。穿著合身的內衣有助防止探針的滑出。 3. 躺在半傾斜的位置感覺較佳。按“+”按鈕以增加電流，直至感到舒服的電刺激，治療期間請放鬆肌肉。
PelvicFloor_P1_enc 療程為期三個月，每天治療1-2次便可。即使症狀消失，也不可終止療程，因為神經再生和血液重建需要時間。以上純粹一般指導。欲更詳細瞭解，我們建議你向骨盆底療法方面的專家查詢。
PelvicFloor_P2 急迫性尿失禁
PelvicFloor_P2_fullname 急迫性尿失禁
PelvicFloor_P2_more 過度活躍的膀胱是由尿道膀胱壁肌肉的偶然收縮所致。過度活躍的膀胱導致突然且無法停止的排尿(尿緊急)，即使膀胱內只存儲少量的尿液。這些自發性的膀胱肌肉收縮可能發生在晚上（夜間發生遺尿的原因之一），但亦有機會在日間發生。膀胱肌肉收縮會經常導致無法預料的少量遺尿。停！如果你感覺小便不正常，請向你的醫生查詢，或參照專家意見，以找到病因。低頻刺激僅有助於膀胱過度活動症，即使在女性陰部神經處於正常活動時，引起膀胱過度活躍的其他因素也會出現。常見引起尿頻的因素有：細菌、濾過性黴菌炎症、腎結石、膀胱頸虛弱。重要！在專家的指導下，可使用EMG（肌電信號）進行骨盆底檢查。鬆弛的肌肉使一滴滴的尿液滲入尿道，形成排尿反射。這和膀胱過度活動症有同樣的感覺，但是可以通過適當的肌肉練習和刺激來治療。研究EMG（肌電信號）能讓你認識到脊椎衰退病變對骨盆底功能的影響。電刺激為改進骨盆底功能機理（神經分佈、血液迴圈、肌肉功能）提供安全，無藥性的治療。治療尿失禁、排尿紊亂的專家，能為你提供合適的檢查，並就合適你的電刺激參數作出建議。在制定治療計畫表，階段性EMG治療及修復進度的評估方面，專家也能提供協助。
PelvicFloor_P2_start 使用說明：\n\n一般規則： \n1. \\>避免食用會引起膀胱過敏的食品及飲料。 \\<\n2. \\>嘗試延遲膀胱排泄的時間。切勿刻意限制液體攝入。\\<\n\n陰道刺激：\n輕輕並穩固地把乾淨且濕潤的電極探針放入陰道。必須確保金屬部分完全放入。探針的金屬表面須朝大腿方向，而其表面須與陰道壁完全接觸。合身的內衣有助防止探針滑出。\n\n按“+/-”鍵，按鍵以調節刺激電流之強度。所使用的電流強弱因人而異，調至感覺舒適便可。勿把電流調至讓自己感到不適的程度，勿使用超過55mA的電流。每天使用1-2次。\n\n功能性刺激：\n刺激骶骨排尿中心，把電極片粘貼在脊椎的任何一側，接近骶骨的位置。能感受到電流的刺激便可，勿使用超過15mA的電流。每天使用2-3次。
PelvicFloor_P2_enc 建議整個療程為期三個月
PelvicFloor_P3 壓力性尿失禁1
PelvicFloor_P3_fullname 壓力性尿失禁1
PelvicFloor_P3_more 壓力性尿失禁是由骨盆神經元和支持膀胱的肌肉受損而造成的，又稱尿失禁。分娩、舉重、咳嗽、便秘、婦科手術、脊椎衰退病變是造成骨盆損傷的典型原因。骨盆肌肉的下垂會同時引致內部器官下垂。當內部膀胱的壓力大於尿道閥的力量時，膀胱所承受的壓力導致尿遺漏。使用EMG機和探針時，請觀察骨盆底如何運作。EMG機除了對治療骨盆底起作用，還有助於治療脊椎衰退病變。
PelvicFloor_P3_start 使用說明： \n1. \\>輕輕並穩固地把乾淨且濕潤的電極探針放入陰道。必須確保金屬部分完全放入。探針的金屬表面須朝大腿方向，而其表面須與陰道壁完全接觸。合身的內衣有助防止探針滑出。 \\<\n2. \\>按“+/-”鍵，按鍵以調節刺激電流之強度，直至感到舒適，所使用的電流強弱因人而異，調至感覺舒適便可。勿把電流調至讓自己感到不適的程度，勿使用超過55mA的電流。 \\<\n3. \\>當感覺到刺激時，試著收縮肌肉；在休息時間段，放鬆肌肉，勿收縮臀部、大腿或下腹。自由呼吸，練習直至感到疲乏。\\<
PelvicFloor_P3_enc 重複使用程式1，每天2次，每週4-5次，直到病情改善。記錄多久一次遺尿、每次多少，觀察改善情況。一旦有了顯著的改善，可使用難一點的程式（有更長的工作時間）進行練習。
PelvicFloor_P4 壓力性尿失禁2
PelvicFloor_P4_fullname 壓力性尿失禁2
PelvicFloor_P4_more 壓力性尿失禁是由骨盆神經元和支持膀胱的肌肉受損而造成的。分娩、舉重、咳嗽、便秘、婦科手術、脊椎衰退病變是造成骨盆損傷的典型原因。骨盆肌肉的下垂引致內部器官膀胱的下垂。當內部膀胱的壓力大於尿道閥的力量時，膀胱所承受的壓力導致尿遺漏。使用EMG機和探針時，請觀察骨盆底如何運作。EMG機除了對治療骨盆底起作用，還有助於治療脊椎衰退病變。
PelvicFloor_P4_start 使用說明： \n1. \\>輕輕並穩固地把乾淨且濕潤的電極探針放入陰道。必須確保金屬部分完全放入。探針的金屬表面須朝大腿方向，而其表面須與陰道壁完全接觸。合身的內衣有助防止探針滑出。 \\<\n2. \\>按“+/-”鍵，按鍵以調節刺激電流之強度，直至感到舒適，所使用的電流強弱因人而異，調至感覺舒適便可。勿把電流調至讓自己感到不適的程度，勿使用超過55mA的電流。 \\<\n3. \\>當感覺到刺激時，試著收縮肌肉；在休息時間段，放鬆肌肉，勿收縮臀部、大腿或下腹。自由呼吸，練習直至感到疲乏。\\<
PelvicFloor_P4_enc 重複使用程式1，每天2次，每週4-5次，直到病情改善。記錄多久一次遺尿、每次多少，觀察改善情況。一旦有了顯著的改善，可使用難一點的程式（有更長的工作時間）進行練習。
PelvicFloor_P5 尿頻/尿急2
PelvicFloor_P5_fullname 尿頻/尿急2
PelvicFloor_P5_more 過度活躍的膀胱是由尿道膀胱壁肌肉的偶然收縮所致。過度活躍的膀胱導致突然且無法停止的排尿(尿緊急)，即使膀胱內只存儲少量的尿液。這些自發性的膀胱肌肉收縮可能發生在晚上（夜間發生遺尿的原因之一），但亦有機會在日間發生。膀胱肌肉收縮會經常導致無法預料的少量遺尿。\n\n停！\n如果你感覺小便不正常，請諮詢醫生，或參照專家意見，以找到病因。低頻電刺激僅對膀胱過動症有幫助。即使女性陰部神經處於正常狀態時，亦有機會出現導致膀胱過動症的其他原因。常見引起尿頻的原因有：細菌、濾過性黴菌炎症、腎結石、膀胱頸虛弱，見下文。\n\n重要！\n在專家的指導下使用EMG（機電信號）進行骨盆底檢查。肌肉鬆弛讓尿液一滴滴滲入尿道，形成排尿反射。這和膀胱過動症的感覺十分相似，可以通過適當的肌肉練習和電刺激來治療。瞭解EMG（肌電信號）能使你認識到脊椎衰退病變對骨盆底功能的影響。\n\n電刺激為改進骨盆底功能機理上（神經分佈、血液迴圈、肌肉功能）提供一種安全、無藥性的治療方法。治療尿失禁和排尿紊亂的專家，能為你提供合適的檢查，並在刺激參數方面，提供建議。他們亦制定治療計畫表及EMG治療修復進度評估方面為患者提供協助。
PelvicFloor_P5_start 使用說明：\n\n一般守則： \n1. \\>避免食用引起膀胱過敏的食物和飲料。 \\<\n2. \\>嘗試延遲排尿時間，切勿限制液體的攝取量。\\<\n\n陰道刺激：\n穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\n\n按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。每天使用1-2次。\n\n功能性刺激：\n刺激骶骨排尿中心。把電極片粘貼在脊椎的任何一側，在骶骨區附近。把電流調至能清楚感覺到的程度便可，切勿使用15mA以上之電流。每天使用2-3次。
PelvicFloor_P5_enc 建議整個療程為期三個月
PelvicFloor_P6 尿頻/尿急3
PelvicFloor_P6_fullname 尿頻/尿急3
PelvicFloor_P6_more 過度活躍的膀胱是由尿道膀胱壁肌肉的偶然收縮所致。過度活躍的膀胱導致突然且無法停止的排尿(尿緊急)，即使膀胱內只存儲少量的尿液。這些自發性的膀胱肌肉收縮可能發生在晚上（夜間發生遺尿的原因之一），但亦有機會在日間發生。膀胱肌肉收縮會經常導致無法預料的少量遺尿。\n\n停！\n如果你感覺小便不正常，請諮詢醫生，或參照專家意見，以找到病因。低頻電刺激僅對膀胱過動症有幫助。即使女性陰部神經處於正常狀態時，亦有機會出現導致膀胱過動症的其他原因。常見引起尿頻的原因有：細菌、濾過性黴菌炎症、腎結石、膀胱頸虛弱，見下文。\n\n重要！\n在專家的指導下使用EMG（機電信號）進行骨盆底檢查。肌肉鬆弛讓尿液一滴滴滲入尿道，形成排尿反射。這和膀胱過動症的感覺十分相似，可以通過適當的肌肉練習和電刺激來治療。瞭解EMG（肌電信號）能使你認識到脊椎衰退病變對骨盆底功能的影響。\n\n電刺激為改進骨盆底功能機理上（神經分佈、血液迴圈、肌肉功能）提供一種安全、無藥性的治療方法。治療尿失禁和排尿紊亂的專家，能為你提供合適的檢查，並在刺激參數方面，提供建議。他們亦制定治療計畫表及EMG治療修復進度評估方面為患者提供協助。
PelvicFloor_P6_start 使用說明：\n\n一般守則： \n1. \\>避免食用引起膀胱過敏的食物和飲料。 \\<\n2. \\>嘗試延遲排尿時間，切勿限制液體的攝取量。\\<\n\n陰道刺激：\n穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\n\n按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。每天使用1-2次。\n\n功能性刺激：\n刺激骶骨排尿中心。把電極片粘貼在脊椎的任何一側，在骶骨區附近。把電流調至能清楚感覺到的程度便可，切勿使用15mA以上之電流。每天使用2-3次。
PelvicFloor_P6_enc 建議整個療程為期三個月。
PelvicFloor_P7 尿頻/尿急4
PelvicFloor_P7_fullname 尿頻/尿急4
PelvicFloor_P7_more 過度活躍的膀胱是由尿道膀胱壁肌肉的偶然收縮所致。過度活躍的膀胱導致突然且無法停止的排尿(尿緊急)，即使膀胱內只存儲少量的尿液。這些自發性的膀胱肌肉收縮可能發生在晚上（夜間發生遺尿的原因之一），但亦有機會在日間發生。膀胱肌肉收縮會經常導致無法預料的少量遺尿。\n\n停！\n如果你感覺小便不正常，請諮詢醫生，或參照專家意見，以找到病因。低頻電刺激僅對膀胱過動症有幫助。即使女性陰部神經處於正常狀態時，亦有機會出現導致膀胱過動症的其他原因。常見引起尿頻的原因有：細菌、濾過性黴菌炎症、腎結石、膀胱頸虛弱，見下文。\n\n重要！\n在專家的指導下使用EMG（機電信號）進行骨盆底檢查。肌肉鬆弛讓尿液一滴滴滲入尿道，形成排尿反射。這和膀胱過動症的感覺十分相似，可以通過適當的肌肉練習和電刺激來治療。瞭解EMG（肌電信號）能使你認識到脊椎衰退病變對骨盆底功能的影響。\n\n電刺激為改進骨盆底功能機理上（神經分佈、血液迴圈、肌肉功能）提供一種安全、無藥性的治療方法。治療尿失禁和排尿紊亂的專家，能為你提供合適的檢查，並在刺激參數方面，提供建議。他們亦制定治療計畫表及EMG治療修復進度評估方面為患者提供協助。
PelvicFloor_P7_start 使用說明：\n\n一般守則： \n1. \\>避免食用引起膀胱過敏的食物和飲料。 \\<\n2. \\>嘗試延遲排尿時間，切勿限制液體的攝取量。\\<\n\n陰道刺激：\n穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\n\n按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。每天使用1-2次。\n\n功能性刺激：\n刺激骶骨排尿中心。把電極片粘貼在脊椎的任何一側，在骶骨區附近。把電流調至能清楚感覺到的程度便可，切勿使用15mA以上之電流。每天使用2-3次。
PelvicFloor_P7_enc 建議整個療程為期三個月。
PelvicFloor_P8 缺乏敏感性
PelvicFloor_P8_fullname 缺乏敏感性
PelvicFloor_P8_more 骨盆肌肉缺乏彈性和肌肉神經系統於生產前後受損是導致陰道敏感性降低和患者不能控制排尿的普遍原因。
PelvicFloor_P8_start 使用說明：\n這是一個多段程式。作為安全預防措施，電流于每段完結前自動減少。如果你感到刺激的水準很低，可按“+”鍵，以增大電流值至先前水準。\n注意：當顯示幕上的電流大於1mA時，才可以改變電流強度。 \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\\<\n2. \\>按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。\\<
PelvicFloor_P8_enc 每天按照步驟治療1-2次，每週5-7次，為約3個月。
PelvicFloor_P9 骨盆鍛煉
PelvicFloor_P9_fullname 骨盆鍛煉-輕微肌肉衰弱的預防或治療法
PelvicFloor_P9_more 骨盆肌肉處於兩腿之間，從前面的恥骨到背部的脊椎末端。它們以吊索的形式承載著你的膀胱和尿道（尿液輸出的管道）。當我們用雙腿走路時，骨盆肌肉在支撐著重要器官，如直腸、膀胱、尿道和陰道上擔當重要的角色。尿道、直腸和陰道是我們天然的排泄口。它們只有在骨盆肌肉處於良好狀態，並能對內腹壓力作出恰當的反應時，才能運作正常。骨盆肌肉的衰弱問題廣泛存在于婦女之中。而在年輕婦女生產後或更年期後的婦女間更為普遍。這與肌肉和年齡老化有關。話雖如此，衰弱肌肉刺激治療對年齡大於80的婦女也產生了奇特的療效。
PelvicFloor_P9_start 使用說明：\n這是一個多段程式。作為安全預防措施，電流于每段完結前自動減少。如果你感到刺激的水準很低，可按“+”鍵，以增大電流值至先前水準。\n注意：當顯示幕上的電流大於1mA時，才可以改變電流強度。 \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\\<\n2. \\>按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。\\<
PelvicFloor_P9_enc 每天按照步驟治療1-2次，每週5-7次，為約3個月。
PelvicFloor_P10 持久力
PelvicFloor_P10_fullname 建立持久力和抵抗疲勞
PelvicFloor_P10_more 建立持久力和抵抗疲勞對需要長期站立和從事體力勞動的女性尤為重要。再者，這個程式對治療混合和壓力性尿失禁十分重要。
PelvicFloor_P10_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。 \\<\n3. \\>當開始感覺到電刺激時，試著收縮骨盆肌肉；在休息時段，試著放鬆肌肉。勿錯誤收縮臀部、大腿或腹部。保持緩慢呼吸，持續練習直到感覺疲勞。\\<
PelvicFloor_P10_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。一旦病情有了顯著的改善，可採用較難的程式（工作時間更長）。
PelvicFloor_P11 放鬆骨盆肌肉
PelvicFloor_P11_fullname 放鬆骨盆肌肉
PelvicFloor_P11_more 慢性忍尿可能導致肌肉長期受壓。亦可能是因為神經受損或供血不正常。這經常導致尿道或腸道排泄困難。其中包括以下後期影響: 肛門裂縫, 骨盆的傳染性炎症等。
PelvicFloor_P11_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。 \\<\n3. \\>當開始感覺到電刺激時，試著收縮骨盆肌肉；在休息時段，試著放鬆肌肉。勿錯誤收縮臀部、大腿或腹部。保持緩慢呼吸，持續練習直到感覺疲勞。\\<
PelvicFloor_P11_enc 每天重複使用程式1-2次，每週4-5次，直至有所改善。記錄治療結果：疼痛減輕、排尿更快速等。如果你注意到骨盆有任何不正常的症狀，請向專家諮詢以獲得進一步治療的建議。
PelvicFloor_P12 分娩后出现遗尿
PelvicFloor_P12_fullname 分娩后出现遗尿
PelvicFloor_P12_more 一般情況下，你的神經，韌帶和盆底肌肉會相互配合來支持膀胱的活動，使尿道緊閉，避免尿遺漏的情況。負擔過重或在妊娠或分娩期間對這些領域造成損傷，可能導致他們無法正常運作。已進行多次生育的母親，尤其是選擇順產的，他們有尿失禁的機會較高。然而，婦女在更年期的時候，不論有否生育過或選擇了哪類型的生產方法，同樣會面對尿失禁這問題，而且機率也差不多。因此保持盆底肌肉的活躍性及健康非常重要。在某些情況下，可能會出現脫垂。在大多數情況下，盆底器官（陰道，尿道，直腸，膀胱）脫垂 ，是盆底肌肉功能嚴重惡化的結果。有時進行外科手術是必要的，繼而配合復康治療。若從一開始便使用此儀器來進行康復訓練，有很大機會避免進行手術。即使有嚴重脫垂（ Ⅱ級）的問題，定期結合EMG和ETS的刺激和自主收縮來進行鍛煉，會有顯著的改善。即使手術被認為是必要的，預先的練習將使手術效果更佳，癒合過程亦更快。
PelvicFloor_P12_start 使用說明： \n在嚴重脫垂或女性器官移位的情況下，我們建議患者同時使用陰道和直腸探頭，以達至更佳效果。\n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道/直腸，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複10次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。 \\<\n6. \\>刺激的第四階段-訓練持久力和耐力，遵從ETS第二階段的電刺激指導。\\<
PelvicFloor_P12_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。一旦病情有了顯著的改善，可採用較難的程式（工作時間更長）。
PelvicFloor_P13 壓力性尿失禁 3
PelvicFloor_P13_fullname 壓力性尿失禁 3
PelvicFloor_P13_more 壓力性尿失禁是由骨盆神經元和支持膀胱的肌肉受損而造成的。分娩、舉重、咳嗽、便秘、婦科手術、脊椎衰退病變是造成骨盆損傷的典型原因。\n骨盆肌肉的下垂引致內部器官膀胱的下垂。當內部膀胱的壓力大於尿道閥的力量時，膀胱所承受的壓力導致尿遺漏。\n使用EMG機和探針時，請觀察骨盆底如何運作。EMG機除了對治療骨盆底起作用，還有助於治療脊椎衰退病變。
PelvicFloor_P13_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複10次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。 \\<\n6. \\>刺激的第四階段-訓練持久力和耐力，遵從ETS第二階段的電刺激指導。\\<\n7. \\>EMG的第五階段跟第一和第三階段相似：請觀察自己是否能更好的控制肌肉，一般來說，肌肉經過熱身後會較易控制。\\<
PelvicFloor_P13_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。一旦病情有了顯著的改善，可採用較難的程式（工作時間更長）。
PelvicFloor_P14 壓力性尿失禁 4
PelvicFloor_P14_fullname 壓力性尿失禁 4
PelvicFloor_P14_more 壓力性尿失禁是由骨盆神經元和支持膀胱的肌肉受損而造成的。分娩、舉重、咳嗽、便秘、婦科手術、脊椎衰退病變是造成骨盆損傷的典型原因。\n骨盆肌肉的下垂引致內部器官膀胱的下垂。當內部膀胱的壓力大於尿道閥的力量時，膀胱所承受的壓力導致尿遺漏。\n使用EMG機和探針時，請觀察骨盆底如何運作。EMG機除了對治療骨盆底起作用，還有助於治療脊椎衰退病變。
PelvicFloor_P14_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複10次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。 \\<\n6. \\>刺激的第四階段-訓練持久力和耐力，遵從ETS第二階段的電刺激指導。\\<\n7. \\>EMG的第五階段跟第一和第三階段相似：請觀察自己是否能更好的控制肌肉，一般來說，肌肉經過熱身後會較易控制。\\<
PelvicFloor_P14_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。
PelvicFloor_P15 急迫性尿失禁 5
PelvicFloor_P15_fullname 急迫性尿失禁 5
PelvicFloor_P15_more 膀胱過動症是由尿道膀胱壁肌肉突然收縮所致，患者會突然立即需要排尿(尿急)，即使膀胱內只有少量的尿液。這些自發的膀胱肌肉收縮可能發生在晚上（引致夜間遺尿的原因之一），但遺尿也會發生在白天。膀胱肌肉收縮經常導致少量遺尿，而且是無法預料的。\n\n停！\n如果你感覺小便不正常，請諮詢醫生，或參照專家意見，以找到病因。低頻電刺激僅對膀胱過動症有幫助。即使女性陰部神經處於正常狀態時，亦有機會出現導致膀胱過動症的其他原因。常見引起尿頻的原因有：細菌、濾過性黴菌炎症、腎結石、膀胱頸虛弱，見下文。\n\n重要！\n在專家的指導下使用EMG（機電信號）進行骨盆底檢查。肌肉鬆弛讓尿液一滴滴滲入尿道，形成排尿反射。這和膀胱過動症的感覺十分相似，可以通過適當的肌肉練習和電刺激來治療。瞭解EMG（肌電信號）能使你認識到脊椎衰退病變對骨盆底功能的影響。\n電刺激為改進骨盆底功能機理上（神經分佈、血液迴圈、肌肉功能）提供一種安全、無藥性的治療方法。\n治療尿失禁和排尿紊亂的專家，能為你提供合適的檢查，並在刺激參數方面，提供建議。他們亦制定治療計畫表及EMG治療修復進度評估方面為患者提供協助。
PelvicFloor_P15_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複10次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。 \\<\n6. \\>刺激的第四階段-訓練持久力和耐力，遵從ETS第二階段的電刺激指導。\\<\n7. \\>EMG的第五階段跟第一和第三階段相似：請觀察自己是否能更好的控制肌肉，一般來說，肌肉經過熱身後會較易控制。\\<
PelvicFloor_P15_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。
PelvicFloor_P16 急迫性尿失禁 6
PelvicFloor_P16_fullname Urge Incontinence 3
PelvicFloor_P16_more 這個程式的設定比程式P15難。一般會於用過程式P15後使用。
PelvicFloor_P16_start 這個程式的設定比程式P15難。一般會於用過程式P15後使用。操作方法已在介紹P15的操作方法時解釋過，請由程式P15開始治療至程式P16。
PelvicFloor_P16_enc 每天重複此程式1-2次，每週4-5次，直至情況改善。記錄多久一次尿失控、每次多少，觀察有什麼變化。
PelvicFloor_P17 急迫性尿失禁 7-修護
PelvicFloor_P17_fullname 急迫性尿失禁 7-修護
PelvicFloor_P17_more 這是治療膀胱過動症（急迫）程式系列中的最後一個。請以先前的程式治療過後並獲得明顯改善，方可使用程式P17。這程式有助肌肉保持良好及穩定的狀態。治療初期，建議使用程式P2來治療嚴重OAB（尿急迫症）。若要進行進一步修復治療，可使用程式P15。
PelvicFloor_P17_start 使用說明： \n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複10次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。\\<
PelvicFloor_P17_enc 若治療停止數周或數月後，發現OAB（尿急迫症）症狀嚴重了，可從程式P15開始治療，每週使用1次，直至感覺改善。如果停止治療數周或數月後，病人狀況是良好，不要增加使用的次數，接著一週一次的方案治療便可。
PelvicFloor_P18 骨盆肌肉鬆弛
PelvicFloor_P18_fullname 骨盆肌肉鬆弛
PelvicFloor_P18_more 肌肉鬆弛也被稱為張力衰退，由於下部神經細胞衰退而降低或減少肌肉的張力。
PelvicFloor_P18_start 陰道探針使用說明：\n穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。\n\n刺激治療：\n1. \\>按“+-”鍵，來調節電流強度，所使用的電流強弱因人而異，只需感覺舒適便可，勿把電流調至感覺不適的程度及電流勿使用超過55mA電流。這是持續的刺激，不是為了肌肉收縮。這個段促進神經傳到性改善和神經肌肉調理。同時創造條件抑制膀胱過度活動。 \\<\n2. \\>如果已達55mA電流仍感覺不到電刺激,應停止治療並且諮詢臨床醫學家。 \\<\n\nETS: \nETS是指自發性的收縮結合電刺激治療。放置好探針並確保輔助線(REF)連接妥當，用以精確地測量EMG資料。試著放鬆至4uv以下，然後按“mA+”鍵來開始ETS治療並調節合適的電流。\n\n一般練習：每天使用這個程式一至兩次，每週4-6天，為期約三個月。持續而有規律的治療有助於優化治療效果。
PelvicFloor_P18_enc 每天重複使用程式2-3次，每週4-6次。最少持續使用三個月，直至感覺有所改善。預約臨床專家，對EMG治療進行評估。
PelvicFloor_P19 骨盆肌肉衰退
PelvicFloor_P19_fullname 骨盆肌肉衰退
PelvicFloor_P19_more 這個程式以ETS為基礎，對治療肌肉痲痹十分有效。 骨盆肌肉處於兩腿之間，從前面的恥骨到背部的脊椎末端。它們以吊索的形式承載著你的膀胱和尿道（尿液輸出的管道）。當我們用雙腿走路時，骨盆肌肉在支撐著重要器官，如直腸、膀胱、尿道和陰道上擔當重要的角色。 尿道、直腸和陰道是我們天然的排泄口。它們只有在骨盆肌肉處於良好狀態，並能對內腹壓力作出恰當的反應時，才能運作正常。 骨盆肌肉的衰弱問題廣泛存在于婦女之中。而在年輕婦女生產後或更年期後的婦女間更為普遍。這與肌肉和年齡老化有關。 話雖如此，衰弱肌肉刺激治療對年齡大於80的婦女也產生了奇特的療效。
PelvicFloor_P19_start 使用說明：\n1. \\>穩固並輕輕地把乾淨且濕潤的電極探針放入陰道，必須確保探針的金屬部分完全放入陰道。探針的金屬表面朝大腿方向，而探針的所有面需與陰道壁接觸。穿著合身的內衣有助於防止探針滑出。 \\<\n2. \\>調節臨界值：EMG臨界值顯示在垂直刻度上。按“AUTO”鍵，盡力收縮肌肉3秒鐘，維持3秒鐘，直至新的臨界值出現。開始使用程式前可隨時重複此動作直至找找到適合自己的臨界值。讓你感覺舒服的臨界值：肌肉收縮時，你的EMG值應該高於臨界值。 \\<\n3. \\>EMG的首個階段-當你看到顯示幕上的向上箭嘴閃爍時，收縮骨盆。當向下箭嘴閃爍時，放鬆。重複十次。 \\<\n4. \\>ETS的第二個階段-使用“+/-”鍵緩緩地調節電流，直至感覺到顯著而舒服的電刺激。勿使用超過55mA的電流。當向上箭嘴閃爍時，收縮骨盆肌肉以達到臨界值，此臨界值是在步驟二時調節好的。到達臨界值時會同時觸發電刺激，說明肌肉維持收縮狀態。當感覺到刺激時，試著自己收縮骨盆。勿收縮臀部、大腿或腹部。緩慢地呼吸，重複15次或直至感到疲乏。 \\<\n5. \\>EMG的第三階段和第一階段相似：請留意自己在進行第二階段治療後，是否更能控制骨盆肌肉。 \\<\n6. \\>刺激的第四階段-訓練持久力和耐力，遵從ETS第二階段的電刺激指導。\\<
PelvicFloor_P19_enc 每天重複使用程式2-3次，每週4-6次。最少持續使用三個月，直至感覺有所改善。預約臨床專家對EMG治療進行評估。
4SportsTile Sports
Sports_P1 運動
Sports_P1_fullname 運動
Sports_P1_more 開始鍛煉前，使用溫和的電刺激，能讓肌肉、心率、血壓和體溫為更強烈的運動做好準備。熱身階段持續30分鐘，包括使用低電流刺激。
Sports_P1_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P1_enc 進行任何訓練之前都需要熱身。
Sports_P2 積極恢復
Sports_P2_fullname 積極恢復
Sports_P2_more 這個程式用於肌肉疲勞後的恢復。過量的乳酸會導致肌肉損傷和疼痛，而這個程式能幫助去除肌肉內的乳酸。
Sports_P2_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P2_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P3 持久性
Sports_P3_fullname 持久性
Sports_P3_more 此程式促進慢氧化型肌肉。目標肌肉負責支援姿勢或任何長期的有氧運動（散步、跑步、和騎車）。
Sports_P3_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P3_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P4 抵抗力1
Sports_P4_fullname 抵抗力1
Sports_P4_more 肌力訓練會增強肌肉力量。同時抵抗疲勞。抵抗力程式1最適合初用者。
Sports_P4_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P4_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P5 抵抗力2
Sports_P5_fullname 抵抗力2
Sports_P5_more 抵抗力訓練會增強肌肉的力量和持久力。抵抗力2程式適用於高級使用者。
Sports_P5_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P5_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P6 抵抗力3
Sports_P6_fullname 抵抗力3
Sports_P6_more 抵抗力训练会增强肌肉的力量和持久力。 抵抗力3程序适用于高级用户。**
Sports_P6_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P6_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P7 抵抗力和肌力訓練1
Sports_P7_fullname 抵抗力和肌力訓練1
Sports_P7_more 肌力和抵抗力訓練能改善肌肉和骨骼的性能。\n訓練的益處：\n-可減輕由肌肉衰弱而引起的背部疼痛。\n- 促進新陳代謝和身體燃燒脂肪的能力。\n抵抗力和肌力訓練程式1提高對初用者極其有效。
Sports_P7_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P7_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P8 抵抗力和肌力訓練2
Sports_P8_fullname 抵抗力和肌力訓練2
Sports_P8_more 肌力和抵抗力訓練能改善肌肉和骨骼的性能。\n訓練的益處：\n-可減輕由肌肉衰弱而引起的背部疼痛。\n- 促進新陳代謝和身體燃燒脂肪的能力。\n抵抗力和肌力訓練程式2對初用者極其有效。
Sports_P8_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P8_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P9 抵抗力和肌力訓練3
Sports_P9_fullname 抵抗力和肌力訓練3
Sports_P9_more 肌力和抵抗力訓練能改善肌肉和骨骼的性能。\n訓練的益處：\n-可減輕由肌肉衰弱而引起的背部疼痛。\n- 促進新陳代謝和身體燃燒脂肪的能力。\n抵抗力和肌力訓練程式3對初用者極其有效。
Sports_P9_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P9_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P10 力量1
Sports_P10_fullname 力量1
Sports_P10_more 此程式通過增加肌肉的體積來提升肌肉收縮時的力量。這使肌肉在限定時間內產生更強的力量。力量1程式適合初級使用者。
Sports_P10_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P10_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P11 力量2
Sports_P11_fullname 力量2
Sports_P11_more 此程式通過增加肌肉的體積來提升肌肉收縮時的力量。這使肌肉在限定時間內產生更強的力量。力量2程式適合高級使用者。
Sports_P11_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P11_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P12 力量3
Sports_P12_fullname 力量3
Sports_P12_more 此程式通過增加肌肉的體積來提升肌肉收縮時的力量。這使肌肉在限定時間內產生更強的力量。力量3程式適合高級使用者。
Sports_P12_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P12_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P13 最大力量 1
Sports_P13_fullname 最大力量 1
Sports_P13_more 此程式促進快肌纖維穩定地建立，這能幫助我們進行以下類型之運動時產生相當大的力量：舉重、格鬥、柔道、球類運動、短跑、騎車等。最大力量程式1適合初用者使用。
Sports_P13_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P13_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P14 最大力量2
Sports_P14_fullname 最大力量2
Sports_P14_more 此程式促進快肌纖維穩定地建立，這能幫助我們進行以下類型之運動時產生相當大的力量：舉重、格鬥、柔道、球類運動、短跑、騎車等。最大力量程式2適合初用者使用。
Sports_P14_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P14_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P15 最大力量3
Sports_P15_fullname 最大力量3
Sports_P15_more 此程式促進快肌纖維穩定地建立，這能幫助我們進行以下類型之運動時產生相當大的力量：舉重、格鬥、柔道、球類運動、短跑、騎車等。最大力量程式3適合初用者使用。
Sports_P15_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P15_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P16 最大力量4
Sports_P16_fullname 最大力量4
Sports_P16_more 此程式促進快肌纖維穩定地建立，這能幫助我們進行以下類型之運動時產生相當大的力量：舉重、格鬥、柔道、球類運動、短跑、騎車等。最大力量程式4適合初用者使用。
Sports_P16_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P16_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P17 爆發力1
Sports_P17_fullname 爆發力1
Sports_P17_more 若肌肉要在短時間內以最佳狀態，產生最大的力量，需要有爆炸力。開始此程式訓練前，要先做持久力和抵抗力程式訓練，以提高承受強度訓練的能力。做好準備功夫方能有足夠的力量完成最後階段的訓練。爆發力程式1適給初級使用者。
Sports_P17_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P17_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P18 爆發力2
Sports_P18_fullname 爆發力2
Sports_P18_more 若肌肉要在短時間內以最佳狀態，產生最大的力量，需要有爆炸力。開始此程式訓練前，要先做持久力和抵抗力程式訓練，以提高承受強度訓練的能力。做好準備功夫方能有足夠的力量完成最後階段的訓練。爆發力程式2適合高級使用者。
Sports_P18_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P18_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P19 對抗肌肉疲勞
Sports_P19_fullname 對抗肌肉疲勞
Sports_P19_more 此程序集中培訓肌肉對抗疲勞。肌肉疲勞表示肌肉產生力量的能力下降。在進行長時間肌肉運動時，肌肉產生力量的的穩定性尤為重要。最常見造成肌肉無力的原因是神經無法長時間維持輸出高頻率信號。肌肉經過一段時間強烈的收縮，神經信號的頻率下降和收縮所產生的力量減小。此時肌肉不再感到疼痛或不適，肌肉不再'聽從'命令，並且逐漸停止活動。高頻肌肉刺激是程序的一部分。它能增加神經的能力來輸出持續，高頻率的信號，使肌肉以其最大的力量收縮，。
Sports_P19_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P19_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P20 對抗肌強化
Sports_P20_fullname 對抗肌強化
Sports_P20_more 對抗肌是我們用於有意識的運動（例如，用二頭肌曲肘、用脛骨肌提腳、用四頭肌把腿由膝蓋伸直，用腹肌彎腰等）。位於骨骼（或脊椎）另一側的對抗肌肉會同時伸展。對抗肌保護一系列的人體動作並且阻止關節移位或受損。哪個是主動肌和哪個主動肌引致某種移動：伸直手臂時，三頭肌成了主動肌（發出動作），二頭肌變成對抗肌（起限制或相反作用-平衡動作）。當一側的均衡肌肉變得衰弱時，微弱的那側應該加強訓練以保護關節和使兩側肌肉力量得以平衡。這個程式適合初級使用者。
Sports_P20_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Sports_P20_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Sports_P21 緩解疼痛
Sports_P21_fullname 緩解疼痛
Sports_P21_more 這個程式用於緩解一般的疼痛。經皮神經電刺激（TENS）提供一種體外無藥性的方法來治療嚴重且長期難治的疼痛。TENS透過電極片把溫和的電刺激傳入皮膚，從而改變身體對痛楚的感覺。TENS不能治癒生理性問題；它僅能幫助控制疼痛感。TENS並不適合每個使用者。尋求建議：請諮詢你的醫生。
Sports_P21_start 使用TENS時，電極片的粘貼位置對能否有效地緩解疼痛是十分重要的。至於哪個位置最合適，請向你的理療師或醫生查詢。亦有可能要把電極片粘到不同的位置進行試驗，直至找到最有效的位置。這些位置可能經過臨近區、皮節區、肌節區、運動點、觸發點或穴位。\n\n皮節區和肌節區由經過中樞神經的單個神經根來支配。每個神經根控制一個皮節區。每個皮節區以控制它的神經根來命名。\n\n臨近粘貼是最常用的電極片粘貼位置。把紅色導線（近身體中央的）放在脊椎旁，也就是皮節（疼痛產生的地方）進入和存在的地方。黑色導線（末梢）通常粘貼在疼痛位置上或其附近。你的理療師或醫生可能讓電流穿過疼痛區，或者使用“bracket”系統，通過疼痛區的神經分支，讓電流穿過疼痛區的任何一側。
Sports_P21_enc 當感到疼痛時，可使用TENS機。治療20分鐘後，如果起不到療效，試著更換電極片的粘貼位置。
4RehabTile Rehab
Rehab_P1 肌肉姿勢控制
Rehab_P1_fullname 肌肉姿勢控制
Rehab_P1_more 這個程式適用於發展緩慢抽搐肌。這類肌肉支持所有類型的基本動作，緩慢的姿勢-如保持某一個姿勢、坐著、手中握東西、站立等。
Rehab_P1_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Rehab_P1_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Rehab_P2 脊椎
Rehab_P2_fullname 脊椎
Rehab_P2_more 這是一種能有效地緩解痛楚的程式。它能治療脊椎神經損傷，一般由於脊椎圓盤退化而導致的，應該把電極片沿著脊柱粘貼，注意，只是在脊柱附近，而不是在其上面。溫和而長期持久的刺激比短而強的電流的刺激更有效。若操作正確，研究證明TENS對修復受損神經是有效的。如果有疑慮，請諮詢理療師。
Rehab_P2_start 1. \\>把電極片粘貼在乾淨、無油脂的皮膚上。電極片沿脊柱粘貼：紅色導線頭連接的電極片應貼在脊柱右側一釐米處，靠近臀部；黑色導線頭連接的電極片應貼在胸椎部上側，離脊柱一釐米處。這樣可讓其中一個通道沿著脊柱約20-60cm。如果儀器有兩個刺激通道，脊柱左側重複以上操作。 \\<\n2. \\>增強電流，直至產生刺痛感，但不要把電流調至讓自己有疼痛或不舒服的感覺。\\<
Rehab_P2_enc 每次可使用此程式兩小時，一旦30分鐘後程式停止運行，按mA鍵重複該程式。
Rehab_P3 四肢輕癱瘓
Rehab_P3_fullname 四肢輕癱瘓
Rehab_P3_more 上下肢肌肉局部癱瘓可能是由於運動神經元受損（中風、神經或脊椎受損、多發性硬化症、腦性癱瘓等）而造成的。這個治療程式可逐漸促進受損後的運動神經和肌肉神經修復。 ETS治療模式是最好的跟進治療模式。
Rehab_P3_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Rehab_P3_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Rehab_P4 面部癱瘓1
Rehab_P4_fullname 面部癱瘓1
Rehab_P4_more 面部癱瘓會擾亂面部活動，例如眼皮下垂、嘴角下垂、臉頰下垂。這個治療程式有助於改善新陳代謝，同時能改變肌肉和皮膚的顏色。\n警告！面部的電極片粘貼位置和在理療師的嚴格指導下方可調節刺激電流。不合適的刺激電流或把電極片粘貼在錯誤的面部或頸部位置，會導致皮膚灼傷和其他嚴重的後果。
Rehab_P4_start 警告！面部的電極片粘貼位置和在理療師的嚴格指導下方可調節刺激電流。不合適的刺激電流或把電極片粘貼在錯誤的面部或頸部位置，會導致皮膚灼傷和其他嚴重的後果。\n1. \\>請使用專業臨床醫生的指導。 \\<\n2. \\>面部電極片有多種尺寸和類型，以刺激面部不同的微小肌肉群。\\<
Rehab_P4_enc 請按規定使用，切勿刺激過度。如果感到灼熱或任何不舒服，請停止使用。由於促進了血液迴圈，面部發紅乃屬正常現象。
Rehab_P5 面部癱瘓2
Rehab_P5_fullname 面部癱瘓2
Rehab_P5_more 面部癱瘓會擾亂面部活動，例如眼皮下垂、嘴角下垂、臉頰下垂。這個治療程式有助於改善新陳代謝，同時能改變肌肉和皮膚的顏色。\n警告！面部的電極片粘貼位置和在理療師的嚴格指導下方可調節刺激電流。不合適的刺激電流或把電極片粘貼在錯誤的面部或頸部位置，會導致皮膚灼傷和其他嚴重的後果。
Rehab_P5_start 警告！面部的電極片粘貼位置和在理療師的嚴格指導下方可調節刺激電流。不合適的刺激電流或把電極片粘貼在錯誤的面部或頸部位置，會導致皮膚灼傷和其他嚴重的後果。\n1. \\>請使用專業臨床醫生的指導。 \\<\n2. \\>面部電極片有多種尺寸和類型，以刺激面部不同的微小肌肉群。\\<
Rehab_P5_enc 請按規定使用，切勿刺激過度。如果感到灼熱或任何不舒服，請停止使用。由於促進了血液迴圈，面部發紅乃屬正常現象。
Rehab_P6 廢用性萎縮
Rehab_P6_fullname 廢用性萎縮
Rehab_P6_more 這個治療程式適用於外科手術或受傷後的肌肉修復。在進行刺激治療之前，你必須得到你的醫務主管的證明，確保你所有傷口和縫針已經癒合。肌肉、肌腱、韌帶受傷後，或其它長期肌肉固化，會導致廢用性萎縮。肌肉刺激可以強化已廢棄的肌肉。
Rehab_P6_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<
Rehab_P6_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Rehab_P7 中風後1
Rehab_P7_fullname 中風後1
Rehab_P7_more 出血性（血管破裂）或缺血性（血塊）中風引起大腦神經元的突然死亡。斷足或者痙攣性癱瘓往往是中風的結果，痲痹，面部痲痹或半邊身痲痹會出現。中風程式1適合初級使用者，直至可以控制一些癱瘓肌肉。請把ETS視作跟進治療模式。
Rehab_P7_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<\n4. \\>當你感覺到電刺激時，試著收縮肌肉，與被動收縮相比主動收縮能更快修復。\\<
Rehab_P7_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Rehab_P8 中風後2
Rehab_P8_fullname 中風後2
Rehab_P8_more 出血性（血管破裂）或缺血性（血塊）中風引起大腦神經元的突然死亡。斷足或者痙攣性癱瘓往往是中風的結果，痲痹，面部痲痹或半邊身痲痹會出現。中風程式1適合初級使用者，直至可以控制一些癱瘓肌肉。請把ETS視作跟進治療模式。
Rehab_P8_start 1. \\>選定要進行電刺激的身體部份，然後把電極片粘貼在乾淨且無油脂的肌肉皮表上。粘貼電極片的一般提示：紅針電極片應該粘貼在肌肉的中心，黑針電極片應粘貼在肌肉末梢（肌肉末梢是指離身體中心點最遠的肌肉[最遠的距離]）。\\<\n2. \\>電極片間的距離至少為1cm:進行電刺激期間，電極片與電極片不能接觸。\\<\n3. \\>可把mA調高至感覺舒適的程度。如果使用超過45mA的電流卻沒有任何感覺，請諮詢臨床專家。\\<\n4. \\>當你感覺到電刺激時，試著收縮肌肉，與被動收縮相比主動收縮能更快修復。\\<
Rehab_P8_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Rehab_P9 多發性硬化
Rehab_P9_fullname 多發性硬化
Rehab_P9_more 多發性硬化（MS）是一種慢性自身免疫混亂性疾病，並且伴隨著多方面的病因。它會影響我們的活動、感知和人體功能。亦會導致神經髓鞘衰退 (神經夾克)，並影響傳達刺激的品質。它能引發以下的疾病: 麻木, 輕度癱瘓, 肌肉僵硬和痙攣, 及尿失禁等。
Rehab_P9_start 這項治療的主要目的是逐漸改善被疾病干擾的神經髓鞘。 \n1. \\>把電極片粘貼在沿神經線路乾淨、無油脂的皮膚上。 \\<\n2. \\>把刺激電流調至感覺舒適的程度便可。若電流已調至45mA卻沒有任何感覺，切勿再加強電流，只需保持10-20mA的低電流便可。\\<
Rehab_P9_enc 每天治療1-2次，每週5天，持續使用3個月。有規律地治療會獲得更好的療效。
Pain_P1 Acute Pain - Gate Ctrl
Pain_P2 Chronic Pain - Opiate
Pain_P3 Chronic Pain - HAN
Pain_P4 Referred Pain - TP
Pain_P5 Opiate - GC - Opiate
Strength_P1 Explosive FT
Strength_P2 Endurance ST
Strength_P3 Stability STAT
Strength_P4 Stability DYN
Strength_P5 Conditioning
Strength_P6 Warm-Up
Strength_P7 WU-Endurance
Strength_P8 WU-Explosive
NMReed_P1 CNS
NMReed_P2 Sensory
NMReed_P3 Repetitive Task Prac.
NMReed_P4 Reciprocal Inhibition
NMReed_P5 Facilitate Recruitment
NMReed_P6 Contralateral Facilitation
UserProt User Protocol
FESswitch_custom Hand Switch Custom
FESswitch_custom_fullname Hand Switch Custom
FESswitch_custom_more FES stands for Functional Electrical Stimulation. It involves using electrical stimulation to activate the weak muscles through the stimulation of intact peripheral motor nerves to promote normal functional movement, for example assisting walking for patients with foot drop. \n\nYou should consult appropriate treatments with a physiotherapist prior to the commencement of treatment. FSE is most effective when highly specific training parameters are selected
FESswitch_custom_start 1. \\>Set up the FES treatment by entering the programme settings screen. Here, you can update the parameters using the on-screen buttons, including the type of stimulation, frequency, pulse width etc.\\<\n2. \\>Tap the Done button to approve the changes.\\<\n3. \\>Connect your electrodes to the lead wire and connect the lead wire to the device.\\<\n4. \\>Using the electrode placement charts, connect the skin electrodes.\\<\n5. \\>Connect the Remote Control to the device\\<\n6. \\>Slowly adjust the intensity with the mA+ and mA- buttons until you reach a suitable level.\\<\n7. \\>Press the hand switch button to activate Manual Control Mode. The intensity will decrease to 0mA and the screen will pass to the Rest Time screen. Here, you will be able to monitor your EMG levels. The Rest time counter will also be displayed.\\<\n8. \\>To start stimulation, press and hold the Remote Switch button (the icon in the header will turn Red). The work time counter shall be displayed in the treatment screen area and stimulation will be delivered at the previously set level.\\<\n9. \\>To stop the stimulation, release the Remote Switch button (the icon in the header will turn White). The device will return to the rest screen, where you will be able to monitor your EMG levels.\\<
FESswitch_custom_enc 
DatabaseVersion_progdata 14
